im电竞投注打开包装后，检查设备是否损坏。 根据结构图，按设备型号安装设备。 应根据装箱单检查配备的部件。 如果加热方式是电加热，请根据工作温度购买相应类型的导热油（注意：导热油绝对不允许含有水分）。 添加时，夹套上部的注油口将打开，夹套上部的油将打开。 这位是开放的。 将油注入加油口，等待油位流过。 油位拧紧后，请勿拧紧燃油口以避免压力。
通过拧紧螺母和螺母来安装阀门，压力表和安全阀，以达到密封效果。 两个接头的电弧密封表面不得相对于彼此旋转。 对于所有螺纹联轴器，应在组装过程中使用润滑剂或润滑油。 石墨避免伤人。 阀门的使用：针阀是密封的，只需要轻轻旋转阀针，紧盖可以达到良好的密封性能，防止过大的力，以免损坏密封面。
反应釜壶体和壶盖的安装和密封：壶体和壶盖与垫圈或锥形表面和弧面接触，主螺母通过按下主螺母拧紧来实现 密封效果好。 拧紧螺母时，须对角对称。 它应该多次重复拧紧，力应该是均匀的。 不要让盖子向一侧倾斜以获得良好的密封效果。 拧紧主螺母时，规定的紧固扭矩不应超过40~120N.M的范围，以防止密封面被压坏或过载。 密封表面应特别小心。 每次安装前，用较软的纸或布擦拭上下密封表面。 请特别注意不要碰到壶体和盖子的密封面。 如果合理操作，可以使用10,000次以上。 在密封表面破裂后，需要重新加工以获得良好的密封性能。 拆卸盖子时，盖子应缓慢上下升降，以防止水壶和盖子之间的密封面相互碰撞。 如果密封件用垫圈（四氟乙烯，铝垫，铜垫，石棉垫等）密封，则可以通过拧紧主螺母来实现良好的密封效果。
Xiaoyi will introduce to you that the electric heating reactor is installed in the high-pressure operating room which meets the explosion-proof requirements.
After opening the package, check whether the equipment is damaged. According to the structure diagram, install the equipment according to the type of equipment. The fittings should be checked according to the packing list. If the heating mode is electric heating, please purchase the corresponding type of heat conducting oil according to the working temperature (note: heat conducting oil is absolutely not allowed to contain moisture). When added, the oil injection port on the upper part of the jacket will be opened, and the oil on the upper part of the jacket will be opened. This is open. Inject the oil into the filling port and wait for the oil level to flow through. After tightening the oil level, do not tighten the fuel nozzle to avoid pressure.
Valves, pressure gauges and safety valves are installed by tightening nuts and nuts to achieve sealing effect. The arc sealing surfaces of the two joints shall not rotate relative to each other. For all threaded couplings, lubricants or lubricants should be used during assembly. Graphite avoids hurting people. Valve use: Needle valve is sealed, only need to gently rotate the needle, tight cover can achieve good sealing performance, prevent excessive force, so as not to damage the sealing surface.
Installation and sealing of kettle body and lid of reaction kettle: The kettle body and lid contact with gasket or conical surface and arc surface, and the main nut tightens by pressing the main nut to achieve good sealing effect. When tightening nuts, they must be diagonally symmetrical. It should be tightened repeatedly and the force should be uniform. Do not let the lid tilt to one side to achieve good sealing effect. When tightening the main nut, the specified tightening torque should not exceed the range of 40-120 N.M. to prevent the sealing surface from being crushed or overloaded. The sealing surface should be carefully sealed. Before each installation, wipe the upper and lower sealing surfaces with softer paper or cloth. Please take special care not to touch the sealing surface of the pot body and lid. If operated properly, it can be used more than 10,000 times. After the sealing surface is broken, it needs to be reprocessed to obtain good sealing performance. When removing the lid, the lid should be lifted up and down slowly to prevent the sealing surface between the kettle and the lid from colliding with each other. If the seals are sealed with gaskets (tetrafluoroethylene, aluminium gasket, copper gasket, asbestos gasket, etc.), good sealing effect can be achieved by tightening the main nut.